Here a simplest yet quite accurate inductance meter is presented, which can be built within few minutes. Furthermore, the circuit can be powered-up with a single 1.5V cell. However, a frequency meter would be required to work-out the inductance.

*Designed and Presented by: Abu-Hafss *

## Using Cross-Coupled NPN BJTs

The circuit is pretty straight forward wherein, two NPN transistors are cross-coupled to form a flip-flop oscillator. The values of R1 and R2 could be anything between 47 - 100R. The frequency of the oscillation is inversely proportional to the inductance and it can be calculated with the following formula:

Frequency (kHz) = 50,000 / Inductance (uH)

### CALIBRATION:

Initially the circuit has to be calibrated using a known inductor, as described below:

Suppose, we have an inductor of 100uH. Putting the value of inductor (100uH) in the above formula, we get 500kHz.

Connect the inductor cross point A & B and power on the circuit. It will start oscillating.

Connect the frequency meter at point A or B and ground.

Adjust the POT until the meter reads 500kHz. Now the circuit is calibrated.

### INDUCTANCE MEASUREMENT:

Connect an unknown inductor across A & B.

Power on the circuit and read the frequency at point A or B.

The above formula can also be written as:

Inductance (uH) = 50, 000 / Frequency (kHz)

Putting the value of frequency in this formula, the value of the inductor can be found.

#### Waveform Image:

Robin says

Hi Abu Hafss

This looks so simple its to good to be true,I'm going to try it.

Where did you derive that formulae from.

Regards Robin

Abu-Hafss says

Hi Robin

Here is the waveform of the oscillation

https://dl.dropboxusercontent.com/u/20969135/Waveform%20Ind%20Meter.png

The frequency of oscillation depends on the L/R time constant comprising the inductor and resistors R1 and R2. The time the waveform takes to change its state is directly proportional to the inductance. In the waveform we see, the time taken for one complete cycle is 2µs for a 100µH inductor.

In other words, inductance (100µH) = Time period (2µs) x 50 ……….(A)

And we know, Frequency = 1 / Time Period OR Time Period = 1 / Frequency

So replacing Time Period in Formula (A), we get

Inductance (µH)

= 50 x (1 / Frequency) OR

= 50 / Frequency (Hz) OR

= 50 x 1000 / Frequency (Hz) x 1000 OR

= 50,000 / Frequency (kHz)

Please note, that this circuit is quite sensitive to the supply voltage and it is fairly good to about 10µH.

It is good idea to a have a known value inductor say 100µH fixed in the circuit for calibration purpose. That calibration inductor and the inductor under test may be toggled using a SPDT switch.

Robin says

Hi Swagatum

I now understand it fully,I built the circuit and calibrated it and tested 3 other known inductance's ie:47uH and got 53.5uH,4.7mH and got 5.9mH,3.9mH and got 4.9mH.

Those figures will give me a good idea of what I've wound.

I also built the other circuit with the 100uA meter which may give more accurate results?But have no idea on how to fault find it and split it into smaller sections so as to check that you are on the right track

Just getting the above results from that tiny little circuit was a big boost for me

Thanks Robin

Swagatam says

Thanks for testing the circuits!

The 555 based design will need different sets of "C" for producing the corresponding sets of measuring range, however the design will not give accurate results with lower capacitance and may show vibrating or oscillating meter needle.

Abu-Hafss says

Hi Robin

Please note that each time the circuit is powered on, it does need to be calibrated. Otherwise, we might get variation in readings.

vishal sutar says

Hi Abu Hafss,

Its a simplest one…thanks for sharing.

I have some question

1. Does the calibration depends on cell voltage.

2. If so then can i give a fixed 5v supply to this circuit.

Abu-Hafss says

Hi Vishal Sutar

1. Yes.

2. Yes.

3. Please take note of my post of Nov. 5, 2014.

Federico says

Hi a question, how did you get the 50000 constant figure in the frequency calculation? Is that a bc546 characteristic? What if I use another npn such as n2222 or others? Does that constant change?

Thanks!

Swag says

Hi, I am not very sure since it was not written by me.

However for a square wave frequency the standard formula seems to be like this IL = Api / 16fL

where IL is the current through the inductor, A is the RMS amplitude of the square wave frequency, pi= 22/7, and f is the applied frequency

Swag says

you can use any similar small signal transistor, it is definitely not critical…2N2222 will work without any issues.

Federico says

Thanks!

Kazeem adesola says

Can we use ordinary multimeter to measure the frequency

Swagatam says

yes that’s possible…

Isaac says

How can we use ordinary multimeter to measure the frequency

Swagatam says

by attaching it to one of these circuits

https://www.homemade-circuits.com/simple-frequency-meter-circuits-analogue-designs/

or buy a multimeter having the frequency measuring feature

Ant says

Hi there, really great forum this thanks for your efforts! I am not very experienced with circuitry and am struggling with how to wire the pot into the circuit… should I have it wired with +volts on pin one, the line that ties gates one and one together to centre pin and ground to pin 3?

Thanks!

Swagatam says

Hi, there are 3 pins in a pot. You will have to join the center pin with one of the outer pins and connect this to ground, and then connect the other free pin of the pot with the emitter pins of the transistor.

Ondrej says

I built this circuit, it works okay for large inductances. It’s important to note though, that it won’t work for small ones, say < 5uH. I wanted to measure 330nH, that would be 150MHz, but the output doesn't go beyond 10MHz. I clearly need a different design for these.

Swagatam says

Thank you for your valuable feedback!