Here a simplest yet quite accurate inductance meter is presented, which can be built within few minutes. Furthermore, the circuit can be powered-up with a single 1.5V cell. However, a frequency meter would be required to work-out the inductance.
Designed and Presented by: Abu-Hafss
Using Cross-Coupled NPN BJTs
The circuit is pretty straight forward wherein, two NPN transistors are cross-coupled to form a flip-flop oscillator. The values of R1 and R2 could be anything between 47 - 100R. The frequency of the oscillation is inversely proportional to the inductance and it can be calculated with the following formula:
Frequency (kHz) = 50,000 / Inductance (uH)
Initially the circuit has to be calibrated using a known inductor, as described below:
Suppose, we have an inductor of 100uH. Putting the value of inductor (100uH) in the above formula, we get 500kHz.
Connect the inductor cross point A & B and power on the circuit. It will start oscillating.
Connect the frequency meter at point A or B and ground.
Adjust the POT until the meter reads 500kHz. Now the circuit is calibrated.
Connect an unknown inductor across A & B.
Power on the circuit and read the frequency at point A or B.
The above formula can also be written as:
Inductance (uH) = 50, 000 / Frequency (kHz)
Putting the value of frequency in this formula, the value of the inductor can be found.
Have Questions? Please Comment below to Solve your Queries! Comments must be Related to the above Topic!!