The proposed Inductance meter circuit can measure inductance or coil values accurately over the given ranges and as a bonus the circuit is also capable of measuring the complimentary capacitor values as accurately.The circuit functioning may be understood with the following points:

As we all know that inductors are fundamentally related to generating frequencies or in other words with pulsating or AC supplies.

Therefore for measuring such components we need to force them with their specific functions in order to enable extraction of their hidden characteristics or attributes.

Here the coil in question is forced to oscillate at a given frequency, and since this frequency depends on the L value of the particular inductor becomes measurable through an analogue device such as a moving coil meter after suitably converting the frequency into amplified voltage/current.

In the shown inductance meter circuit, T1 along Lo, Lx, Co, Cx together forms a Colpitts oscillator type of self oscillating configuration, whose frequency is directly determined by the above L and C components.

Transistor T2 and the associated parts help amplifying the generated pulses at the collector of T1 to reasonable potentials which is fed to the next stage comprising T4/T5 for further processing.

The T4/T5 stage raise the current and integrate the acquired info to appreciable levels so that it becomes readable over the connected uA meter.

Here Cx and Co basically provides the range selection option, many good quality caps with precise values may be positioned in the slot, with a provision of selecting the desired one via a rotary switch. This will allow an instant selecting facility of any desired range for enabling wider measurement of any particular inductor.

Conversely, correctly measured inductors/capacitor may be positioned at Co, Lo and Lx for getting an equivalent meter deflections for any unknown capacitor at Cx.

P1 and P2 may be used for monitoring and adjusting the zero position of the meter, it also allows fine tuning of the selected range over the meter.

Meter FSD calibration can be achieved by using the formula: ni = nm(1 – fr)/(1 – fc) where ni is the number of divisions measured on the scale, nm = total number of division of the scale, fr = relative frequency, fc = the smallest relative frequency measured.

The current consumption would be around 12mA at 12V while an inductor is being measured.

Robin says

Hi Swagatam

I've made a function generator and tried to pass different wave forms through inductors to try and work out the inductances but no positive results.

I want to try and understand the above circuit.

I know how to work out the resonance frequency in the colpitts oscillator(two caps and one inductor in the tank circuit)

How do you include that in the above diagram.

Is it,If you know the value of Co,Cx,Lo then you can work out Lx somehow?

…….If you know the value of Lo,Lx, and Co,then can you work out Cx?

And the deflection on the ammeter how would you change it to farads or henries

Regards Robin

Swagatam says

Hi Robin,

waveform will have no effect on inductors r capacitors so it cannot be used as a parameter for determining inductance or capacitance values.

Yes in the above diagram all the mentioned elements are interdependent, basically one must have a standard inductor and capacitor with known values, then these can be used for calibrating the meter. Once the meter gets calibrated with these known values, other unknown values can be proportionately calibrated over the meter dial ….since the reading would be absolutely linear.

Soumyadip Mishra says

Sir,

What is the configuration of the meter? Can I use a PMMC type voltmeter or other?

Swagatam says

yes, a moving coil type meter will be required for this design, but a 100uA is the specified rating, a V meter will need to be upgraded or modified appropriately for making it compatible

Sunny Yadav says

R8,R11,R14,R1 in rasistains ki valu kiya hi saf dikh nahi raha ro y 10uH my kitny tarns hi y Lo,Lx kiya hi bataiy sir

Swagatam says

R8 = 560 ohms

R11 = 5.6k

R14 = 5.6k

R1 = 680 ohms

L0 is a fixed inductor, Lx is the one which needs to be tested.

10uH inductor can be procured readymade from any electronic retailer shop

riyaz says

What software do you use for simulating and verifying these designs? thx.

Swagatam says

mind simulation